Test Guidelines for Medical Gas regulators, flowmeters, suction valves
(AS 3840, AS 2902, AS 2120.3 AS 2473.3)
Gas regulators, flow-meters, and suction valves, which are intended to be connected pressure regulators for the administration of medical gases, must meet the specification requirements of the Australian Standard. Confirmation that these devices are performing
accurately, reliably and within manufacturer’s specifications is a requirement for users of medical gases.
U-Tech Medical supports this Standard as can provide assistance and guidance to adopt the best management program, specifically for the devices types in your medical facility.
How we can help?
- We can review your list of equipment and provide you with a recommendation for the specific tests required and the frequency of testing for each device.
- With our calibrated test equipment we can provide a validation report indicating compliance with this standard.
- We can advise on the condition of your devices, and recommendation on the life cycle of each device so you can plan a replacement program before the equipment becomes non-serviceable.
Frequently asked questions and answers
Q. How often does my equipment need to be tested?
A. The assessment should include, manufacturers recommended intervals, impact and likelihood of failure, experience with similar equipment, level of usage, statutory requirement, and service history. Typically, it is industry standard to test devices at least annually.
Q. What tests are normally performed during service?
A. Physical damage, testing of zero point, leak test, regulator performance test, pressure relief valve operation, flow across the range. Seals and filters are replaced on an annual basis, and regulator seat on a 3 year basis.
Q. What happens if my equipment fails the test?
A. U-Tech Medical will advise the corrective action and the cost if repair is needed.
Q. Do I need to purchase a copy of the Australia Standards?
A. This is not required, however if you prefer to keep a copy, it is available on-line at: http://infostore.saiglobal.com/store/ and type “AS/NZS 3840.1” in the search window. Other standards may also be downloaded by typing the required code.
Q. How do I select the best biomedical service company for my equipment management plan?
A. There are a number of factors to consider when choosing the correct service company. U-Tech Medical has developed a “Service Provider Evaluation Checklist” which can be downloaded from our homepage or the top of this page or simply ask one our customer service team members on 1300 4 UTECH, (1300 488324) to send you the report.